Portable Emergency Dental Phlegm Suction Unit Medical Vacuum Aspirator Machine. High negative pressure, large flux, low noise. Safe to Use: The machine does not produce a positive pressure to ensure that its performance is reliable, safe to use. 1000ML Water Tank: This item reliefs in less than a minute, 1000ml high capacity water tank for less refills. Portable: Small size, light weight, easy to carry, especially to adapt to a variety of emergency situations and travel needs.
The limit negative pressure:= 0.07MPa. Negative pressure adjustment range:0.02MPa to the limit negative pressure. Pumping rate:= 15L / min.
Fuse tube:AC250V, F1.5AL f5 × 20. Vacuum gauge: 00.1MPa, level 2.5. Equipment type: high vacuum/low flow.
Dimensions: 29cm15cm28cm/11.41"5.9"11.02. Inhale the water before sucking to moisten the inner wall of the suction tube and the latex tube to prevent the sputum from adhering to the inner wall of the tube and causing cleaning trouble.
The end of the suction tube was wetted with physiological saline before sucking and then sucked. Do not take more than 15 seconds each time, if you use it again, it should be separated for 3 minutes to prevent damage to the visceral mucosa. If the patient's visceral mucosa is inadvertently sucked during sucking, the suction tube should not be taken out at will.The suction tube should be opened in time to release the negative pressure. After the patient's discomfort disappears, the suction tube can be taken out.