Photo-Activated Disinfection technology is based on two components to selectively tag and destroy cariogenic bacteria and periodontal pathogens: a solution of dilute, pharmaceutical grade tolonium chloride (a vital stain) and light of a specific wavelength (650 nm) to activate the solution. Photo-Activated Disinfection(PAD) Light-F3WW provide the light of a specific wavelength (650 nm) to activate the solution.
Introduced into dental hard or soft tissue, the solution of dilute, pharmaceutical grade tolonium chloride (a vital stain) selectively targets and tags all bacteria. When the solution is activated by the Photo-Activated Disinfection(PAD) Light-F3WW, it releases singlet oxygen which ruptures the cell walls of bacteria, killing them in seconds. 1.2 Indications for Use. Photo-Activated Disinfection(PAD) Light-F3WW is equally effective to activate the solution on the bacteria found in root canals, periodontal pockets, periimplantitis, carious teeth, Infection control and pain relief.Kills all bacteria including: Streptococcus mutans Total streptococcus Streptococcus sobrinus Streptococcus intermedius Actinomyces Lactobacillus veilonella Prevotella intermedia Peptostreptococcus micros Fusobacterium nucleatum Porphyromonas gingivalis Staphaloccoccus aureus E. 2.2 CLEANING AND STERILIZATION BEFORE USE ON PATIENTS. The PAD Light and accessories are provided non-sterile. Cleaning and sterilization of accessories and components that come in contact with open tissue is required prior to use on patients. Do not stare into the visible light. Laser light is emitted from this device.
Do not point the device at anyones eyes. The device will need to be fully charged prior to the first use. Plug the AC recharging power cord into the unit and then into the wall socket. Charge for at least 12 hours before the first use.
Recharge the device at least once every 3 weeks whether you have used the device or not. Put battery into the cylinder of the the charge station (Contact PCB faces down), the green light turns on when its charging, the light will turn off when the battery get full charge. Its about 2 hour to get full charge. Push the Power/Select Button to power on the light. Push the Power/Select Button to select the seconds for the procedure.Push the Light Button to activate the Light. Push the Light Button to stop the light. When the light is activated, there is a bee tone in every 30seconds. Note:The light will be turned off after 3 mins without any operation.
Pull out the battery to stop the power. Because the device contains a laser, do not shine it directly into your eyes or look directly at the laser light. Do not point the PAD Light device at your or another persons eyes. The safety of laser devices on pregnant women has not been established.
Avoid use of the PAD Light device over, or in the area of, cancerous lesions. The device should not be used during the following activities: bathing or showering sleeping driving or operating machinery. The following is recommended: Do not drop the device or handle it roughly. Do not immerse the device in water or any liquid. If using the device on more than one person, sterilization of light rob for each use.
Length without a light rod. Diameter of the out put of the light rod. Continuous model of the laser beam. The sale of this item may be subject to regulation by the U.Food and Drug Administration and state and local regulatory agencies. The item "Medical Dental Photo Activated Disinfection Heal Diode Oral Laser PAD Light" is in sale since Wednesday, August 21, 2019. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\Dental Lasers".baistra" and is located in Shenzhen.
This item can be shipped worldwide.