1.2 Indications for Use Photo-Activated Disinfection(PAD) Light-F3WW is equally effective to activate the solution on the bacteria found in root canals, periodontal pockets, periimplantitis, carious teeth, Infection control and pain relief. Kills all bacteria including: Streptococcus mutans Total streptococcus Streptococcus sobrinus Streptococcus intermedius Actinomyces Lactobacillus veilonella Prevotella intermedia Peptostreptococcus micros Fusobacterium nucleatum Porphyromonas gingivalis Staphaloccoccus aureus E.2.2 CLEANING AND STERILIZATION BEFORE USE ON PATIENTS The PAD Light and accessories are provided non-sterile. Cleaning and sterilization of accessories and components that come in contact with open tissue is required prior to use on patients. 2.3 Safety Do not stare into the visible light. Laser light is emitted from this device. Do not point the device at anyones eyes.
2.4 BATTERY CHARGING The device will need to be fully charged prior to the first use. Plug the AC recharging power cord into the unit and then into the wall socket. Charge for at least 12 hours before the first use.Recharge the device at least once every 3 weeks whether you have used the device or not. Put battery into the cylinder of the the charge station (Contact PCB faces down), the green light turns on when its charging, the light will turn off when the battery get full charge. Its about 2 hour to get full charge. Push the Power/Select Button to power on the light. Push the Power/Select Button to select the seconds for the procedure.
Push the Light Button to activate the Light. Push the Light Button to stop the light. When the light is activated, there is a bee tone in every 30seconds.Note:The light will be turned off after 3 mins without any operation. Pull out the battery to stop the power. WARNINGS Because the device contains a laser, do not shine it directly into your eyes or look directly at the laser light. Do not point the PAD Light device at your or another persons eyes. The safety of laser devices on pregnant women has not been established.
Avoid use of the PAD Light device over, or in the area of, cancerous lesions. The device should not be used during the following activities: bathing or showering sleeping driving or operating machinery 5. The following is recommended: Do not drop the device or handle it roughly. Do not immerse the device in water or any liquid.
If using the device on more than one person, sterilization of light rob for each use. Length without a light rod. Diameter of the out put of the light rod. Continuous model of the laser beam.
The sale of this item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies. The item has been cleaned and handled in accordance with the manufacturers instructions. If the item you received is damaged or defective, Please tell us immidately.
Our guarantee does not cover accidental damage or wear of product caused by customer. Thank you very much for your understanding!
You feedback is VERY important to us. We understand the concerns and frustrations you might have, and will try our best to resolve the issues. Service is our #1 priority at BAISTRA! If you tell us immediately after placing the order, we will make every effort to assist you with your change request. We offer the best items at the best prices.We strive to exceed all your expectations and put a smile on your face. We stand behind our products and pride ourselves on OUTSTANDING customer service. Trade/Device Name: LED Curing Light Regulation Number: 21 CFR 872.6070 Regulation Name: Ultraviolet Activator for Polymerization Regulatory Class: Class II Product Code: EBZ Dated: December 7, 2019 Received: December 16, 2019. The item "Dental PAD Photo-Activated Disinfection Medical Light Lamp Laser" is in sale since Thursday, February 25, 2021. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\Dental Lasers". The seller is "baistramed" and is located in Shenzhen. This item can be shipped worldwide.