Open the air releasing valve to discharge cold air inside the sterilizer. When the air releasing valve start going out a little hot steam, close the air releasing valve. When the steam pressure reaches 0.142-0.165MPa , the safety valve will start pressure relief work, in order to maintain a certain pressure. After sterilization, unplug the power cord. 3 minutes later, open the air releasing valve, when the pressure gauges to zero, open the lid to discharge the remaining air.Safety Measure and Maintenance 1. This sterilizer is steam pressure vessel so do not impact in using and forbid overpressure. If the pressure pointer of pressure gauge reaches the 0.165MPa and the safety valve does not exhaust, it should stop using at once and check up the safety valve. The pointer should be at zero position before usage the pressure gauge, otherwise the gauge has been failure and it should repair or replace the gauge. The pressure gauge should be regulated 12 times per year, ensuring sensitiveness and correctness.
Gasket is easy to wear. If it does not wok well, see it whether become aging and hard, then replace it in time.
The item "18L Professional Medical Steam Autoclave Sterilizer Dental Lab Equipment" is in sale since Thursday, June 11, 2020. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\Autoclaves & Sterilizers". The seller is "lavental" and is located in Altadena, California. This item can be shipped to United States, Canada.