Chamber capacity:ø25×D28CM (ø9.84×D11.02inch). Outside Size :46.53536 CM. Auto supply water system: Manual give the water. Safety valve: automatic discharge in excess of standard value. Temp and Press display:Pressure gauge.
Disinfection principle:Uses the vapor power emissions cold air / 2 times. Plug in the power cord. Shut exhaust needle valve in the Close (clockwise)°. Adding about 800 ml of distilled water.
Close the door cover, tight the rotate handle. Start the timer knob to 121? Sterilization is complete, please be sure to open the exhaust needle valve firstly (inverse clock direction). Wait for the pressure gauge pointer to zero, opened the door cover.Every time sterilization, must add about 800 ml of distilled water. Use a separate socket (confirm the socket of the output voltage).
Pressure gauge pointer in the red areas, please be sure to first Open the exhaust steam of the needle valve. Plate 3pcs (Size:20113621 mm). Take Plate tool 1pc steam exhaust pipe 1pc(1 meter length). Tank capacity: 18L Inside dimensions: f250 x D350mm Sterilize temperature: 122? Sterilize pressure: 0.12Mpa & 0.23Mpa Sterilize time: 25min & 6min Drying time: 40min Tank Material: Stainless steel #304 Machine size (LWH): 60 39 36cm Packing size (LWH): 61 53 40cm N.35KG Standard accessories: (Plate 3PCS; Shelf 1PCS; Plate Taking tool 1PC) Application: For dental, surgery, beauty salon, medical care, laboratory, and so on. After the completion of disinfection, please to wait for 10 minutes. Pressure Gauge in red area. Do not rotate handle, open the steam discharge valve firstly.
Before each use, must supply water 800ml of distilled water. Please close steam discharge valve before the disinfection. Do not use any corrosive or chemical water as the disinfection. Do not share the same electricoutllet with other electrical apparetus. Please pay attention to correct voltage and frequency.
Connect steam exhaust pipe with the steam vent, before operation. Chamber capacity:ø25×D35CM (ø9.84×D13.78inch). Temp and Press display :Led several windows. Uses the vapor power emissions cold air / 2 times.Before each use, distilled water is better. If you are planning on combining your items, please make sure that you wait until ALL auctions have ended before you visit the checkout. All items must be paid within 7 days. You feedback is VERY important to us!
We understand the concerns and frustrations you might have, and will try our best to resolve the isuues. Exchanges are offered for items that are defective or damaged when received, although this is highly unlikely. Defective items or problems with your order must be reported within 3 days after receiving. Please keep all packaging, documentation, packing list, etc.That came with your order. These will be used to verify claims if there is any problem with you. Serivce is our #1 priority at Dentistclub! We offer the best items at the best prices. We strive to exceed all your expectations and put a smile on your face. We stand behind our products and pride ourselves on OUTSTANDING customer service. The sale of this item may be subject to regulation by the U.
Food and Drug Administration and state and local regulatory agencies. K181993 Device Classification Name Sterilizer, Steam 510(K) Number K181993 Device Name STURDY Autoclave Super Microm (Models SA-260MA And SA-260MA-R) Regulation Number 880.6880 Classification Product Code FLE Date Received 7/26/2018 Decision Date 11/1/2019 Device Class 2 Regulation Number 21CFR880.6880. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\Autoclaves & Sterilizers".The seller is "dentistclub" and is located in this country: US. This item can be shipped to United States.