14L Dental Steam Autoclave Sterilizer Medical Sterilization S/S 304 Lab Description. Chamber capacity:ø25×D28CM (ø9.84×D11.02inch) Inside capacity:14 Liter Material:Stainless steel #304 Outside Size :46.53536 CM Packing Size :513940 CM G. W:15KGS Sterilizer temperature:121? Sterilize pressure:0.12Mpa Sterilize time:25Min Power:900W AC110V/8A AC220V/4A Auto supply water system: Manual give the water Safety valve: automatic discharge in excess of standard value Temp and Press display:Pressure gauge Disinfection principle:Uses the vapor power emissions cold air / 2 times.Plug in the power cord 2. Shut exhaust needle valve in the Close (clockwise)° 3. Adding about 800 ml of distilled water. Close the door cover, tight the rotate handle 4.
Start the timer knob to 121? Sterilization is complete, please be sure to open the exhaust needle valve firstly (inverse clock direction) 6. Wait for the pressure gauge pointer to zero, opened the door cover.
Every time sterilization, must add about 800 ml of distilled water Use a separate socket (confirm the socket of the output voltage) Pressure gauge pointer in the red areas, please be sure to first Open the exhaust steam of the needle valve. Main unit 1pc Standard accessories: Plate 3pcs (Size:2115.52 cm) Shelf 1pc Take Plate tool 1pc steam exhaust pipe 1pc(1 meter length). 18L Dental Autoclave Steam Sterilizer Medical Sterilization with Drying Function. Inside dimensions: f250 x D350mm.Sterilize pressure: 0.12Mpa & 0.23Mpa. Sterilize time: 25min & 6min. Tank Material: Stainless steel #304. Machine size (LWH): 60 39 36cm. Packing size (LWH): 61 53 40cm. Standard accessories: (Plate 3PCS; Shelf 1PC; Plate Taking tool 1PC). Application: For dental, surgery, beauty salon, medical care, laboratory, and so on. After the completion of disinfection, please to wait for 10 minutes. Pressure Gauge in red area.
Do not rotate handle, open the steam discharge valve firstly. Before each use, must supply water 800ml of distilled water. Please close steam discharge valve before the disinfection. Do not use any corrosive or chemical water as the disinfection. Do not share the same electricoutllet with other electrical apparetus.Please pay attention to correct voltage and frequency. Connect steam exhaust pipe with the steam vent, before operation. The sale of this item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Handpieces & Instruments\Dental Handpieces".
The seller is "picusmall" and is located in this country: US. This item can be shipped to United States.