Inner dimensions: 500 x 120 x 175 mm. Carton dimensions: 515 x 220 x 180 mm.
Plug the power cord into an AC outlet and adjust the timer according to thickness of the material to be sealed. Use a high-number for thicker material. In order to enhance the quality of the sealing, perform the resistance pre-heating. Press the sealer arm twice simulating the sealing without a package.
This procedure may be performed only once before starting the sealing procedure itself but it should be repeated every time the equipment goes idle for longer than 1 minute. Place the envelope on the sealing platform always with the paper side down (directly on the resistance) and the plastic side up and press the sealer arm down firmly. The heating time is controlled automatically after manual adjustment. When the red LED goes off the sealing will be completed.
Slide the blade to cut the envelope, raise the sealer arm and remove the package. A In order to make the sealing better, wait for 1 or 2 seconds to release the sealer arm after the red LED goes off. B If the heating time is too long or insufficient the sealing might be inefficient (the envelope may be stuck to the sealing area when it is pulled or be visibly badly-sealed).C To correct this, adjust the timer properly. D Once adjusted it is not necessary to perform any other adjustment to continue sealing. Keep the sealing platform always clean. The accumulation of waste on the sealing area will reduce the life cycle of the resistance and the insulating material. For an effective cleaning, wait for the sealing area to cool down and use a cloth slightly moistened with alcohol. If the insulating material lower tape is worn the sealer may go faulty and the resistance may be damaged.
Being so, every time the resistance is replaced, please also check the insulating material tape and replace it if it is worn. If it is necessary to replace the resistance or the fuse, use only those which have the same specifications as the genuine parts'. While packing the instruments protect the ones which have cutting and piercing tips with cotton or gauze to avoid the packages rupture. Place the sealed envelopes with instruments into the autoclave.
Replace the balde after cutting around 800 times. Use pliers to take out the obtuse one, replace the new blade. Replace heat-resisting fabric after sealing about 500 times or inefficient seal.Extract the silicon tube, peel off the heat-resisting fabric, replace the new one, and inset the silicon tube at original position. The sale of this item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies. Please make sure it's correct. Don't worry, I will tell you before you pay if it's need. Please check the package first before you sign for the package. Thanks for your cooperation 2. Our guarantee does not cover accidental damage or wear of product caused by customer.
Thank you very much for your understanding! You feedback is VERY important to us! We understand the concerns and frustrations you might have, and will try our best to resolve the issues. Moredental has been a leading based dental supplies, majored in Dental Equipment and Dental Instruments Sales.
Some customers can't believe how cheap we are and the good quality of the products! Our version is to offer the best dental instrument at the best prices to satisfy our customers' needs. We cooperate with more than 19 top brands, like Woodpecker, JINME, Tosi, and Denjoy. We ensure to provide you best price with best quality. You will be replied within 24 hours(48 hours in the weekend).The item "Dental Sealing Machine Autoclave Sterilization Sealer for Medical Home Food Use" is in sale since Wednesday, June 1, 2016. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\Autoclaves & Sterilizers". The seller is "moredental" and is located in Shanghai.
This item can be shipped to North, South, or Latin America, all countries in Europe, all countries in continental Asia, Australia.