Nm to activate the solution. Provide the light of a specific wavelength (650 nm) to activate the solution.
Introduced into dental hard or soft tissue. Of dilute, pharmaceutical grade tolonium chloride (a vital stain).Selectively targets and tags all bacteria. It releases singlet oxygen which. Ruptures the cell walls of bacteria, killing them in seconds. 1.2 Indications for Use. Found in root canals, periodontal pockets, periimplantitis. CLEANING AND STERILIZATION BEFORE USE ON PATIENTS. And accessories are provided non-sterile. Cleaning and sterilization of accessories and components that come in contact with open tissue is required prior to use on patients. Do not stare into the visible light. Laser light is emitted from this device. Do not point the device at anyone. The device will need to be fully charged prior to the first use. Plug the AC recharging power cord into the unit and then into the wall socket. Charge for at least 12 hours before the first use.
Recharge the device at least once every 3 weeks whether you have used the device or not. Put battery into the cylinder of the the charge station (Contact PCB faces down), the green light turns on when it. S charging, the light will turn off when the battery get full charge.
S about 2 hour to get full charge. To power on the light.
To select the seconds for the procedure. When the light is activated, there is a.Note:The light will be turned off after 3 mins without any operation. Pull out the battery to stop the power. Because the device contains a laser, do not shine it directly into your eyes or look directly at the laser light. Device at your or another person. The safety of laser devices on pregnant women has not been established. Device over, or in the area of, cancerous lesions. The device should not be used during the following activities. Like any electronic product, the. Do not drop the device or handle it roughly. Do not immerse the device in water or any liquid. If using the device on more than one person, sterilization of light rob for each use.
Diameter of the out put of the light. Continuous model of the laser beam.
The sale of this item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies. IMPORTANT INFORMATION ON OUR POLICY AND FEEDBACK. Feedback if you are satisfied with our items. We try to be fair and accurate with all of our product listings and descriptions.
If there is a problem, we can work together to resolve it. We strive to always be extremely fair and always try to work with reasonable customers.
We will always make every attempt possible to resolve issues if we made a mistake. Please do not assume a mistake is intentional. Let us know if our service could be better! Tracking info of your parcel is available.
The following form is for your reference. Please repack the item carefully.510(K) Number: K181691 Regulation Number: 872.4200 Classification Product Code: EFB Subsequent Product Code: EGS. The item "Dental Heal Laser Diode PAD Photo-Activated Disinfection Medical Light Lamp HOTK" is in sale since Monday, April 10, 2017. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\Dental Lasers". The seller is "dentistskysea" and is located in Beijing; China. This item can be shipped worldwide.