12L Dental Autoclave Sterilizer Medical Vacuum Steam Sterilization Class N. LCD display, user friendly interface. Unique touch panel with easy clean surfaces. New slide safe lock design, easier to open & close. Short cycle and high volume pre and post vacuum pump for perfect sterilization result.
Efficient safety systems, pressure release valve, computerized diagnosing, alarm system. New design with easy cleaning surfaces 8. 6 cycles of sterilization 11.
16 security systems Specifications 1. Power input: AC110V/220V; Power output: 1200W 3. Frequency: 50 / 60Hz 4. Sterilization programs: 1 program 5. Deaeration and Depressurization: Automatic 7.Chamber dimensions: 220 x 300mm 8. Autoclave dimensions: 335 x 330 x 485mm 9. Weight: Aluminum:19.4kg(including components) Stainless stell : 21.2kg(including components) 11. Overall clearance: 10cm for each side of the autoclave 14. Clearance required for the movement of the door: 34cm 15. Clearance between the trays: 67mm(approximately) 16. Dimensions of the trays: 155x12x280mm(small)/200x12280mm(large) 17. Maximum and minimum steam pressure: 0 to 4kgf/cm2 19. Temperature of drained water: 100 degree 20. Total heat in joules transmitted in one hour: 15 degree to 40 degree 21.
Proper working alttitude: up to 3500m Possible sterilization failures: 1. Presence of residual air in the chamber and/or in the package: 2. Packages are too big and heavy; 3. Inappropriate wrappings(material composition) for sterilization in steam autoclaves; 4.Insufficient exposure time to the sterilizing agent; 5. Obstruction of the internal orifices and /or steam exit of the autoclave dueto the lack of daily cleaning; 7. The autoclave can be loaded up to 75% of its chamber capacity, that means 5 envelopes for Vitale Plus 12 or 13 envelopes when using the support. The use measures should be standardized for all cycles and offices. For this standardization see" Sterilization Monitoring Process". Tearing/puncturing of the packages while loading or removing them from the autoclave; 9. Lack of autoclaves's preventive maintenance; 10. The chosen cycles is not suitable for the article to be sterilized; 11. Equipment failure, it must be observed by the operator during the cycle. Do not allow patients or especially children to get close to the autoclave. Install the autoclave in an exclusive sterilization room.
Never warm up or sterilize any kind of food in the autoclave. Never make any kind of experiment with animals in the autoclave.
Never make any kind of use of this equipment other than the ones describled in this manual. South America, Argentina, Brazil, Mexico, Chile, Ecuador guar, Uruguay.The sale of this item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies. EUB (under 2KG): Only for USA, Canada, and Australia. E-Express(under 30kg and Non-battery): Only for Japan, Korea, Singapore, Taiwan, Australia, UK, France, Holland, Canada, Russia, Brazil, Ukraine, Belarus, Spain.
Few states need to pay remote surcharge. Don't worry, I will tell you before you pay if it's need. Please make sure it's correct.Your SATISFACTION is our ultimate goal. We work hard to earn your trust and future business. A Please check the package first before you sign for the package. B If you have a defective item, you can return it or ask for a discount. Or replace item for you. We are factory founded dental equipment Wholesaler and Retailer.
If you have any questions about our service or products. You will be replied within 24 hours(48 hours in the weekend).
The item "12L Dental Autoclave Sterilizer Medical Vacuum Steam Sterilization Class N" is in sale since Saturday, September 28, 2019. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\Autoclaves & Sterilizers". The seller is "goden1314" and is located in Beijing.
This item can be shipped worldwide.